This is an old revision of the document!
This tool is designed to be as quick and easy to use as possible to help to refine the mount polar alignment.
To start the procedure select the menu Tools → Polar alignment
Before to start this tool the mount must be aligned with the pole the best as possible.
The camera must be connected and if possible the mount must be initialized to allow to automatically slew at the required position.
Point the telescope near the pole, at a declination between 89° and 90°. It is important the pole is not exactly at the image center.
The camera use the Slew setting exposure, binning and filter. Be sure this setting give a fast enough image refresh.
In the first screen, select if the mount can slew automatically. This generally require the handpad is initialized and a quick star alignment was done.
Select if the mount will move West or East direction, and the movement between two measurement. The mount will move two time this amount, so be sure there is no obstacle in the path.
If the mount cannot slew automatically, because it is a non-goto mount or the handpad is not initialized you can select to move the mount manually.
When ready click the Start button.
The next screen show the progress of the measurement, take exposure, plate solve, move to next position.
If you select the automatic telescope move you have nothing to do until this step is complete.
In the case of manual move, it will stop at each “Rotate the telescope” step.
Move the telescope only along the RA axis by about 15° to 45°
It is very important to not change the declination and to do the two rotation in the same direction.
But it is not important if the amount of rotation is not the same for the two rotation.
You can move the telescope using the handpad or manually by losing the RA axis knob. When the telescope is stable in the new position you can click the Continue button.
After all the steps are complete it change to the final screen.
This show the result of the computation with the current mount axis direction and the total polar alignment error. Be careful at the value after the +/- sign, if it is not zero this indicate an inconsistency in the measurement.
Then it show the correction to do on the mount in both the horizontal and vertical axis.
The direction are to be interpreted this way:
At the same time the camera start a loop of exposure and an overlay is show on the image.
Move the green circle at an extremity of the line to a star you can easily recognize later, for example a star that is part of a small figure, or the brightest star on the image.
You can zoom on the image with the mouse wheel to help to center the star. Press the Ctrl key if you need to move the image instead of the overlay.
When the overlay is in place, click the button Lock overlay, this way you can zoom and move the image without the risk to move the overlay.
Now the goal is to move the star to the pink circle at the other extremity of the line by using only the mount polar axis adjustment knob.
During all this procedure you can change the exposure time of the preview but do not change the geometry of the image, no change of binning or ROI!
After the star reach the pink circle you can close this screen, the alignment procedure is complete.
In the case the alignment error is bigger than the camera sensor size you cannot show the pink circle on the image. In this case move the star to the point the green line exit the image field. Then redo the full procedure until the polar error is small enough to show the two circle.